The RHM 80 excels in tough applications where high temperatures, pressures and corrosive fluids are present. Its heavy duty construction withstands extreme applications without losing Coriolis meter performance.
- Range: 80 kg/min up to 8000 kg/min
- Pressure: up to 299 bar (4336 psi)
- Temperature: -196°C (-321°F) up to +350°C (662°F)
- Accuracy: 0.1 % of rate
- Materials: SS 316L / SS 316Ti, Super Duplex – 1.4410, Alloy C22 – 2.4602
- Process Connection: 6" up to 6"
Nominal Flow (Qₙₒₘ)*
5000 kg/min (11023 lb/min)
Maximum Flow (Qₘₐₓ)*
8000 kg/min (17637 lb/min)
Typical Minimum Flow (Qₘᵢₙ)*
100 kg/min (176 lb/min)
Fluid temperature range options cover applications from -196°C to +350°C (-320°F to +662°F). For integral transmitter versions please refer to transmitter datasheet
-50°C to 80°C (-58°F to 176°F)
Up to 341 bar / 4946 psi - dependent upon material
Electrical Connection Sensor
M20 x 1.5 standard cable entry for JM, SM terminal box versions
Optional entries available : ½” NPT or M25 x 1.5 (only for SM) or ¾” NPT (only for SM)
Max. cable length to remote RHE transmitter 100 m / 328 ft
Sensor Enclosure Materials
Stainless steel 304 (standard), SS 316 (optional)
Coated aluminum terminal box, SS 316 terminal box (optional)
Protection class IP65, NEMA 4 (standard); IP66 / NEMA 4X (optional)
1.4571 (SS 316Ti), 1.4410 (SuperDuplex) - standard
2.4602 (Alloy C22) - consult Factory
Additional/customer specific materials available upon request
Nearly any – the Rheonik AnyPipeFit Commitment. Consult factory for types/sizes not listed in this data sheet on the Mechanical Construction pages
Pressure Rating Compliance
Europe – PED: Module A2, Module B3.1+C2
Canada – CRN: Canadian Registration Number
Certifications and Approvals
ATEX / IECEx Approvals for Zone 0, 1 (details see page 10)
North American Approvals Class I, Div. 1, 2, Gr. A,B,C,D, Zone 0, 1, 2
MID custody transfer approval (OIML R117)
American Bureau of Shipping (ABS) Product Type Approval for use on marine vessels
Testing and Inspection
All sensors are hydro tested, calibrated and supplied with a traceable calibration certificate. Customized calibration and testing services are available
Project Documentation and QA, Services
Rheonik offers a full set of services for large and complex engineering projects.
•Certificates of origin and conformity, mill certificates
•Data books including WPAR, WQS, NDT, test & quality plans, functional testing, calibration procedures, customized packing, factory acceptance etc.
•Painting to project specification
•Start up and commissioning services on/offshore
Enclosure heating for high temperature applications
Cleaning for oxygen service , …
For more consult factory
* At Qₙₒₘ pressure drop across a parallel tube sensor will be approximately 0.4 bar (6 psi) for H₂0. Sensors can be operated at higher flow rates up to Qₘₐₓ but pressure drop will be higher. Typical Minimum Flow Qₘᵢₙ is the recommended lowest flow rate for an accurate measurement. Sensors will measure flow rates lower than Qₘᵢₙ but uncertainty will increase beyond 1 % of rate.
The flow rate specifications above relate to standard pressure, parallel tube, manifold sensor versions. Models with higher pressure ratings have increased wall thickness and will have higher pressure drops.
At Qnom, pressure drop across a parallel tube
sensor will be approximately 1-3 bar (15-44 psi)
for H20.Sensors can be operated at higher flow rates but pressure drop will be higher. Qmin is
the recommended lowest flow rate. Sensors will measure flow rates lower than Qmin, but uncertainty will increase beyond 0.5 % of rate.
The flow specifications above relate to standard pressure, parallel tube, manifold sensor versions. Models with higher pressure ratings have increased wall thickness and will have higher pressure drops.